II. Improved generator4. Enlarged mm resection loop1. Safe bipolar technology Need a big reason to switch to the new. GYNECARE VERSAPOINT ™ II. Central Sterile · Surgical Instruments · Endoscopic Devices · Hysterscopes. Item Shown. GYNECARE VERSAPOINT Bipolar Electrosurgery System by Ethicon. MAUDE Adverse Event Report: ETHICON GYNECARE VERSAPOINT During hysteroscopy, the versapoint resectoscope was in the patient but not in use.
|Published (Last):||5 February 2008|
|PDF File Size:||18.7 Mb|
|ePub File Size:||7.79 Mb|
|Price:||Free* [*Free Regsitration Required]|
It will not damage paint or leave sticky residue.
Add to My Bench. Hand piece is wet, or electrode has touched hysteroscope during activation.
Used GYNECARE Versapoint Electrosurgical Unit For Sale – DOTmed Listing #
Service manual Electric Instantaneous Water Heaters Service manual Before installation quoting stage Work out with the customer what is the right model to install for the application Versapount assessment 1. Check inlet fuse and replace with the correct type if necessary. After replacing brake pads, depress brake pedal firmly More information.
Minimally invasive gynecological procedure Operative Hysteroscope or resectoscope is inserted through the cervix into the uterine cavity Endoscopic surgical instruments are used to remove pathology Allows surgeons to see and treat in a single outpatient intervention 4.
Ethicon Gynecare Versapoint Resectoscopic System Electrode (Each)
To ensure that your installation proceeds smoothly–please Gersapoint information. Flush handle assembly D. Item Qty Description Service More information. Although the install looks daunting at first, once you get the learning curve Vefsapoint information. Service Manual Model No. Do not insert sharp or tiny More information. Check insulation of connector cable.
The driver must depress the brake pedal in order to move the gear selector from Park. Fill in pertinent information on page for future reference. What Happens Before Your More information.
Fill in pertinent information on page 3 for future reference. Plastic Replacement Sleeve 2. Ensure no contact was made with other equipment during activation. Purpose The purpose of this document is to describe the procedure necessary to replace one of gynecaree ball nuts More information. Ethicon Versapoint User Manual. PUB Revision B The names of any providers and patients used in illustrations or examples in this document are fictitious.
Ensure approved footswitch is attached. Check grounding of generator.
You can add Community Subscriptions in the search bar that says “Subscribe to more communities Forums Documents Parts Videos News. Start display at page:. Before installation, ensure camshafts are aligned, crankshaft is locked in place and camshaft gear bolts are loose as gynecaare in removal procedures. Remove the U-bolt nuts More information. Index These step by step how to manuals will give you the foundation necessary to use your new LED lights.
It is disclosed to the receiving. Check integrity of electrode. Allow generator to cool down before re-use. Unauthorized duplication More information. Be careful not to scratch or hit front edge of the side viewing micro prisms onto hard objects!
Attach the light cable verwapoint the light source to the hysteroscope light post. Cable and Fiber Preparation Attach the assembled standard obturator and inner sheath to the outer sheath Insert the assembled standard obturator and inner sheath assembly into the outer sheath Screw the inner sheath into the outer sheath using the ribbed ring on the outer sheath Inner Sheath Outer Sheath Gynscare Accessing the printer system board Accessing the printer system board 1 Turn the printer off.
Disconnect hoses and carefully gyynecare for an obstruction. You can adjust your Community Subscriptions in Settings.
Attach the standard obturator to the inner sheath. Troubleshooting Guide and Instructions for Service http: Engage the latch pawl. Safety precautions 3 B.